Repair and Calibration Services
8565E spectrum analyzer 9kHz-50GHz
Our full repair facility can repair and calibrate your Agilent 8565E, HP 8565E Spectrum Analyzer. Analyzer Repair Inc. specializes in the Agilent 8565E and HP 8565E. Analyzers are our business..
We are specialists in the repair of the Agilent/HP 8565E spectrum analyzers. The Agilent 8565E spectrum analyzer offers the measurement capability and performance previously found in larger, more expensive benchtop analyzers. Agilent/HP 8565E full repair support.
Frequency accuracy: ± 1 kHz (CF = 10 GHz). Frequency Span Range: 0 Hz, 100 Hz to maximum frequency. Frequency Readout Accuracy Span>2 MHz x N: ± (freq. readout x freq. ref. accuracy +5% x span +15% x RBW +10 Hz). Frequency Readout Accuracy Span =2 MHz x N: ± (freq. readout x freq. ref. accuracy +1% x span +15% x RBW +10 Hz). Sweep Time Range: Span = 0 Hz: 50 µs to 6,000 s; Span =100 Hz: 50 ms to 100 ks. Resolution bandwidth range: 1 kHz to 2 MHz. Resolution bandwidth Accuracy: 1 Hz to 300 kHz: ±10%; 1 MHz: ±25%; 2 MHz: +50%, -25%. Video Bandwidth Range: 1 Hz to 3 MHz in a 1, 3, 10 sequence. Amplitude Range: Displayed average noise level to +30 dBm. Average noise level (narrowest RBW): -143 dBm. Optimum dynamic range 2nd/3rd order: 117 dB / 100 dB. Amplitude accuracy: ± 3dB.
001, Second IF Output (310.7 MHz). 005, Alternate Sweep Out (cannot be used with Option 002). 006, 30 Hz to Freq. Coverage. 007, Digitized Fast Time-Domain Sweeps. 008, Signal Identification. 042 Protective Soft Carrying Case/Backpack. 103, Delete Precision Frequency Reference, and 1 Hz and 3 Hz RBWs. 104, Delete Mass Memory Module.
Located in: Keysight (Agilent HP)