Repair and Calibration Services
8563E spectrum analyzer 9kHz-26.5GHz
Our full repair facility can repair and calibrate your Agilent 8563E, HP 8563E Spectrum Analyzer. Analyzer Repair Inc. specializes in the Agilent 8563E and HP 8563E. Analyzers are our business..
We are specialists in the repair of the Agilent/HP 8563E spectrum analyzers.The Agilent 8563E portable, color display RF spectrum analyzer offers the measurement capability and performance previously found in larger, more expensive benchtop analyzers.
Frequency range: 9 kHz to 26.5 GHz; 30 Hz to 26.5 GHz (Opt. 006). Frequency accuracy: ± 1.2 kHz (CF = 10 GHz). Frequency Span Range: 0 Hz, 100 Hz to maximum frequency. Frequency Readout Accuracy Span >2 MHz x N: ± (freq. readout x freq. ref. accuracy +5% x span +15% x RBW +10 Hz). Frequency Readout Accuracy Span =2 MHz x N: ± (freq. readout x freq. ref. accuracy +1% x span +15% x RBW +10 Hz). Sweep Time Range: Span = 0 Hz: 50 µs to 6,000 s; Span =100 Hz: 50 ms to 100 ks. Resolution bandwidth range: 30 Hz to 3 MHz. Resolution bandwidth Accuracy: 1 Hz to 300 kHz: ±10%; 1 MHz: ±25%; 2 MHz: +50%, –25%. Video Bandwidth Range: 1 Hz to 3 MHz in a 1, 3, 10 sequence. Amplitude Range: Displayed average noise level to +30 dBm. Average noise level (narrowest RBW): -117 dBm. Optimum dynamic range 2nd/3rd order: 103 dB / 81 dB. Amplitude accuracy: ± 2.7dB.
Opt 001 - Second IF Output (310.7 MHz). Opt 005 - Alternate Sweep Out (cannot be used with Option 002). Opt 006 - 30 Hz to Freq. Coverage. Opt 007 - Digitized Fast Time-Domain Sweeps . Opt 008 - Signal Identification. Opt 026 - APC-3.5 mm Input Connector. Opt 103 - Delete Precision Frequency Reference, and 1 Hz and 3 Hz RBWs. Opt 104 - Delete Mass Memory Module.
Located in: Keysight (Agilent HP)